Electro Surgical Diathermy : D450V1
Electro Surgical Generator
MODELS – D450V Series represent D450V1
Tissue sensing technology.
Two monopolar outputs.
Endo cut i.e. pulsed cut for cutting polyps.
High-Cut mode for cutting fatty area with ease.
Various coag modes viz. Fulgurate, Soft, Force and Spray.
Power and Mode control through hand switch
Safety features like Patient Plate Contact Monitoring, HF leakage monitoring, Arc control, Auto Stop
There are 8 modes in AUTO CUT:
4 CUT effects and 4 HIGH CUT effects. HIGH CUT mode is especially useful for tissues with poor electric conduction, such as fatty tissue while CUT mode is
recommended for delicate tissues EFFECTS 1 to 4 give different haemostasis levels. The ENDO CUT program is specially designed for cutting polypectomy and papillotomy. ENDO CUT is an electrosurgical cut which is carried out in
predefined intervals. The ENDO CUT function regulates cutting speed and haemostasis. In this way, a polypectomy loop cannot cut through a polyp at just any speed.
Forced coagulation is the standard coagulation mode.Fulgurate coagulation provides the cut as well as coagulation effect at the same time. This reduces the surgery time. Soft (also known as dessicate) coagulation is contact coagulation with deep effect. This treats tissues gently and reduces the necrosis & tissue adhesion to the electrode.Spray coagulation is ideal for noncontact superficial coagulation.
Bipolar Cut can be activated with touch of a button by switching to ‘b’program.
Bipolar coagulation can be operated in manual (activation and deactivation by footswitch) or in automatic modes. In Semi-Auto mode, the activation is manual and the deactivation takes places once coagulation reaches optimum level.In Auto mode, both activation and deactivation are automatic.
You can personalize the power setting and different modes, your all settings
will be stored in memory. There are 10 Programs. Power fail memory: All current settings stored at the time of power fail and are instantly available on switch on of the machine. Automatic self diagnostic startup check will detect common failures like short circuited foot switches, short circuited hand switches and also its own power supply. The errors detected are stored in memory and are useful for service engineer.
The 2+1 pedal footswitch makes it
possible to use bipolar coagulation
and monopolar coagulation without
change of any settings on the
machine. It is also possible to use 2
Gynaecology and Obstetrics
Laparascopy and Cystoscopy
Urology including TUR
Plastic surgery & Dermatology
Monopolar cut (Auto Cut Effect 4) :300W in 500 Ohms (340 KHz+/-15%)
Haemostasis levels (EFFECTS) : Four
Forced coag. power :120W in 350 Ohms (Pulse Modulated)
Spray coagulation power :120W in 500 Ohms (Pulse Modulated)
Fulgurate power :120W in 500 Ohms (Pulse Modulated)
Soft Coag. Power: 120W in 125 ohms (340 KHz +/-15%)
Auto Bipolar output power
Precise and Macro Mode :120W in 125 Ohms (350 KHz +/-15%)
Micro Mode :80W in 125 Ohms (350 KHz +/-15%)
Bipolar Cut: 150W in 500 Ohms (350 KHz +/-15%)
Power display accuracy : +/-15% OR 5W whichever is greater
Mains supply : 230V (+/- 15%) 50Hz (+/- 5Hz)
Internal power supply: SMPS
Maximum current :4.0 A
Power consumption in standby mode :30W
Operating temperature range : +10 to +40 °C
Dimensions :W 420 x H 155 x D 350 mm
Weight: 7.2 Kg.
Disposable Hand Switch 1No.
Patient Plate Silicon Split
Bipolar Forcep – Yes
Foot Switch 2 Pedal+1 Pedal
- Order in The Next to get it by June 1, 2023
- Real Time 126 Visitor Right Now
Electro Surgical Diathermy : D450V1